4-Pole sensors were designed for the measurement of liquid conductivity. The engineering design based on 4-electrode principle guarantees excellent linearity from 1 μS/cm up to 300 mS/cm. The main application for Conducell sensors is the in-line measurement under demanding conditions found in biotechnology and pharmaceutical applications.
|Specification||Conducell 4UxF||Conducell 4US|
|Measuring Range||1 µS/cm - 300 mS/cm||0.1 µS/cm - 500 mS/cm|
|Operating Temperature Range||-20 – 150 °C
(Arc: analog 0 – 110 °C, digital 0 – 130 °C)
|-20 – 135 °C|
|Pressure Range (bar g)||-1 – 20 bar (135 °C)
-1 - 10 bar (140 °C)
|0 – 6 bar|
|Hygienic Aspects||Autoclavable, CIP, SIP||CIP, SIP|
|Electrode Material||S=Stainless Steel 1.4435
H=Stainless Steel 2.4602
|Stainless Steel 1.4435|
App Note - Arc DO, pH, Cond in production scale fermentation process
App Note - Arc sensors in oncologic and biosimilar applicationsInnoxtorres, a Spanish bioreactor manufacturer took advantage of Arc sensor technology to integrate EasyFerm Plus and VisiFerm DO sensors directly into their control system.
App Note - Arc DO, Cond in monitoring of heating & cooling systemsHevasure, a sister company of Midland Corrosion Systems, protects HVAC assets by continuous monitoring dissolved oxygen and conductivity in the heat transfer liquids at various customers in UK.
Brochure - BioPharma
Catalog - Process AnalyticsDownload the Process Analytics sensor catalog to learn about the complete offering of Hamilton process sensors and accessories.
Brochure - Arc Intelligent SensorsArc Intelligent sensors offer a new way to connect to your process control system.
Brochure - BreweryLearn about our control and analytical solutions that are essential to the brewing process.
Can Hamilton conductivity sensors be used with my existing transmitter?
Does temperature affect conductivity measurement?Yes, temperature has a direct impact on conductivity. Increasing temperature increases the mobility of ions within a solution and results in higher conductivity measurements. Temperature compensation can be performed using empirically derived temperature coefficients, which may be different for various solutions.
Which liquid conductivity standard should I use for calibration?A conductivity standard should be chosen that most closely matches the anticipated conductivity range of the process.
What can I use to clean the conductivity sensor electrodes if fouling occurs?The metal electrodes of 2 and 4-pole sensors can be cleaned with water and mild detergents and washed with deionized water after cleaning. Care should be taken to not use cleaning agents that can impact the material of the electrodes. Click to reference the knowledgebase article on cleaning for more information.
Conductivity Sensor Selection
We're Ready to Help
Hamilton's sales, customer service, and technical support staff are available for questions and additional information about all products and services.Contact Us