Needle Gauge Chart
When selecting needle gauge sizes it is important to keep in mind the volume of the syringe and the dead volume of the needle. For example, it will be very difficult to prime a 10 µL syringe if the dead volume in the needle is greater than 10 µL. Refer to the needle gauge chart to choose a needle gauge with an appropriate µL/inch before selecting a needle. Select the minimum needle length that allows you to carry out your application comfortably.
Note: The ‘s’ on a 22s needle represents a smaller I.D. (inner diameter) for the needle and a thicker needle wall for better durability. For example, a 26 gauge needle has an O.D. (outer diameter) of 0.46 mm and an I.D. of 0.26 mm while the 26s gauge needle has an O.D. of 0.47 mm and an I.D. of 0.13 mm. The 26s has half the I.D. of the 26 gauge needle. Also, the difference in the wall thickness nearly doubles with 26s gauge having a thickness of 0.18 mm while the 26 gauge is only 0.10 mm.
|Gauge Number||Needle Nominal O.D.|
|Needle Nominal I.D.|
|Needle Wall Thickness|
|Needle Dead Volume|
|34 gauge needle||0.159||0.051||0.051||0.052 µL/25.4 mm|
|33 gauge needle||0.21||0.108||0.051||0.233 µL/25.4 mm|
|32 gauge needle
||0.235||0.108||0.064||0.233 µL/25.4 mm|
|31 gauge needle||0.261||0.133||0.064||0.353 µL/25.4 mm|
|30 gauge needle||0.312||0.159||0.076||0.504 µL/25.4 mm|
|29 gauge needle||0.337||0.184||0.076||0.675 µL/25.4mm|
|28 gauge needle||0.362||0.184||0.089||0.675 µL/25.4 mm|
|27 gauge needle||0.413||0.21||0.102||0.876 µL/25.4 mm|
|26s gauge needle||0.474||0.127||0.178||0.322 µL/25.4 mm|
|26 gauge needle||0.464||0.26||0.102||1.349 µL/25.4 mm|
|25s gauge needle||0.515||0.153||0.178||0.464 µL/25.4 mm|
|25 gauge needle||0.515||0.26||0.127||1.349 µL/25.4 mm|
|24 gauge needle||0.566||0.311||0.127||1.93 µL/25.4 mm|
|23s gauge needle||0.642||0.116||0.267||0.268 µL/25.4 mm|
|23 gauge needle||0.642||0.337||0.152||2.266 µL/25.4 mm|
|22s gauge needle||0.718||0.168||0.279||0.563 µL/25.4 mm|
|22 gauge needle||0.718||0.413||0.152||3.403 µL/25.4 mm|
|21 gauge needle||0.819||0.514||0.152||5.271 µL/25.4 mm|
|20 gauge needle||0.908||0.603||0.152||7.255 µL/25.4 mm|
|19 gauge needle||1.067||0.686||0.191||9.389 µL/25.4 mm|
|18 gauge needle||1.27||0.838||0.216||14.011 µL/25.4 mm|
|17 gauge needle||1.473||1.067||0.203||22.715 µL/25.4 mm|
|16 gauge needle||1.651||1.194||0.229||28.444 µL/25.4 mm|
|15 gauge needle||1.829||1.372||0.229||37.529 µL/25.4 mm|
|14 gauge needle||2.109||1.6||0.254||51.076 µL/25.4 mm|
|13 gauge needle||2.413||1.804||0.305||64.895 µL/25.4 mm|
|12 gauge needle||2.769||2.159||0.305||93 µL/25.4 mm|
|11 gauge needle||3.048||2.388||0.33||113.728 µL/25.4 mm|
|10 gauge needle||3.404||2.693||0.356||144.641 µL/25.4 mm|
|*mm are nominal|
Looking for Information not found in our Syringe Needle Gauge Chart?
Check out our page for Hamilton Laboratory Products
See the Hamilton line of Syringe Needles
Try the Visual Needle Builder
Take the hassle out of syringe needle selection with Hamilton's visual tool to guide you through the selection process.
Hamilton Values Higher Education
Hamilton is committed to higher education and believes in the importance of developing the next generation of scientists. Learn how you can save 25% on all syringes, needles, and HPLC columns purchased throughout the year.
Syringes for Higher Education
For over 60 years Hamilton has been committed to supporting academic research and innovation with syringe grants and special discounts. Help your grant money stretch further.
What is a Hamilton Syringe Product Grant?
Hamilton Company proudly supports higher education by awarding up to five $1,000 Syringe Product Grants each month to teaching facilities that use Hamilton syringes or needles as educational guides. In addition to $1,000 worth of Hamilton syringes you will also receive our Syringe Care and Use Guide to be used as an educational guide.