Demanding applications can leach the electrolyte from the pH sensor causing measurement inconsistency and eventual failure. Hamilton refillable pH sensors solve this problem by allowing the user to replenish the electrolyte.
WHAT MAKES REFILLABLE SENSORS DIFFERENT?
Replenish The electrolyte
Processes like pure water, leach electrolyte from the sensor. Sensors like the IonoTrode allow manual refilling without removing the sensor, saving downtime and replacement costs.
Up the pressure, up the life
By using external pressure, the ChemoTrode can deliver consistent electrolyte flow. Process chemicals that would normally poison or clog the sensor are displaced. The design solves critical applications in the chemical, petrochemical, and oil / gas industry.
Great for continuous applications
Refillable sensors work well in continuous measurement applications such as flowing pipelines or flowcells. The sensor can be isolated through armature for service or simply serviced online.
|MEASURING RANGE||0 - 14 pH||± 2000 mV||0 - 14 pH||0 - 14 pH|
|PROCESS TEMPERATURE||0 – 130 °C||0 - 130 °C||-10 – 40 °C||0 – 130 °C|
|PRESSURE RANGE||0 – 6 bar||0 - 6 bar||0 – 0.5 bar (higher|
with pressurized sidearm)
|0 – 6 bar|
|HYGIENIC ASPECTS||CIP, SIP||None||None||SIP|
|SENSING ELEMENT||PHI Glass||Pt ring||F Glass||PHI Glass|
|ELECTROYLTE||Viscous 3M KCl-LR|
(Bridge - Skylyte,
P - Protelyte)
|Viscous 3M KCl-LR||3M KCl||Skylyte|
(Bridge - Everef-B,
P - Everef-F)
(Bridge - Platinum,
P - HP Ceramic)
|HP Ceramic||Sleeve||HP Coatramic|
Catalog - Process Analytics
Manual - Process pH and ORP - TraditionalGeneral Process pH and ORP sensor manual for EasyFerm, Polilyte Plus, ChemoTrode, FermoTrode, IonoTrode, MecoTrode, Polilyte Pro, Polyplast Pro, Liq-Glass, and EasyControl products.
EU Declaration of Conformity - Analog Sensors and ArmaturesEU Declaration of Conformity to the RoHS 2 directive 2011/65/EU and ATEX Directive 2014/34/EU. The declaration applies to the product families: pH sensors, dissolved oxygen sensors, conductivity sensors, armatures and fittings.
EU Declaration of Conformity - Analog Sensors and Accessories (non ATEX)EU Declaration of Conformity to the RoHS 2 directive 2011/65/EU. It applies to all "non ATEX" products or product families such as: laboratory electrodes and cables, process sensors and cables, housings and accessories.
What is a refillable sensor and why would I choose it?Refillable sensors have a port that allows manual replenishment of the reference electrolyte. Processes like pure water leach out out the electrolyte so the Ionotrode is a good choice. Other products like the ChemoTrode allow the for externally pressurized electrolyte so the process liquid cannot easily contaminate the sensor.
How do I clean an ORP sensor?The platinum band around the tip of the ORP sensor can oxidize over time. Light scrubbing with a mild abrasive such as toothpaste will remove the oxidization. Rinse with potable water and store in storage solution prior to usage
What is the difference between ORP and Redox?They are the same. Redox is a shortened word for “Reduction Potential”. ORP is shortened for “Oxidation Reduction Potential”. The sensor and measurement is the same.
How to Measure Sensor A-lengthSensor A-length determines the insertion depth of the sensor. This article shows how this measurement is made for various styles of sensors.
Powered vs. non-powered cablesSome sensors require standard cables while other sensors require power. Which cable should you use? This quick guide helps better answer these questions.
pH and ORP Sensor Installation TipsThis brief article describes some common considerations with sensor installation and removal, as well as best practices for mounting.
We're Ready to Help
Hamilton's sales, customer service, and technical support staff are available for questions and additional information about all products and services.Contact Us