Hamilton is one of the few sensor providers to make all of our own liquid calibration standards, buffers, and electrolytes in-house. These products are specifically designed to work with our sensor products lines and ensure the highest quality measurement along with full traceability of the reference material during future calibration or maintenance work.
Cell Density Solutions
DuraCal pH Buffers
DuraCal ORP Buffers
What Makes DuraCal pH and ORP Buffers Different?
pH and ORP liquid buffer solutions can easily be overlooked when it comes to maintaining measurement accuracy. Hamilton DuraCal buffers mitigate these risk of calibration errors through innovative formulation and bottle design. Learn more with the video below.
Brochure - Liquid Solutions for Calibration
Catalog - Process AnalyticsDownload the Process Analytics sensor catalog to learn about the complete offering of Hamilton process sensors and accessories.
Brochure - BioPharmaInnovative solutions for process analytics in biotechnological and pharmaceutical industries.
How does the DuraCal buffer bottle work?
How often do Incyte sensors (pre-amp version) need to be calibrated?Hamilton recommends once per year with a validation at 6 months.
Which liquid conductivity standard should I use for calibration?A conductivity standard should be chosen that most closely matches the anticipated conductivity range of the process.
What can I use to clean the conductivity sensor electrodes if fouling occurs?The metal electrodes of 2 and 4-pole sensors can be cleaned with water and mild detergents and washed with deionized water after cleaning. Care should be taken to not use cleaning agents that can impact the material of the electrodes. Click to reference the knowledgebase article on cleaning for more information.
Stability and Accuracy of DuraCal pH Buffers
Temperature Influence on Conductivity StandardsTemperature has a direct effect on the conductivity of liquid standards. This page lists the temperature effects on all Hamilton standards in one easy to use place.
Stability and Accuracy of Conductivity StandardsQuestions often come up regarding liquid conductivity standards and how these products should be used. This knowledgebase article takes on many of these topics.
Documentation and Certificates for Sensors and Liquid SolutionsWhat documentation and certificates can I expect with my sensor or probe? This article provides examples of the types of paper documents that normally come with Hamilton products.
We're Ready to Help
Hamilton's sales, customer service, and technical support staff are available for questions and additional information about all products and services.Contact Us