The controller features a large, easy-to-use touchscreen with a processor more than 20 times faster than the original controller. A dedicated host USB port enables connection to a keyboard, mouse, printer, and barcode reader
1. Memory Slot
2. Optional Power Input
3. Host USB for Mouse, Keyboard, Printer, and Barcode Scanner Ethernet
4. Slave USB Port
5. Optional RS-232
- Simple Diluting — Diluent is drawn in from the reservoir by the left syringe and sample is drawn into the hand probe by the right syringe. Both syringes dispense through the hand probe to complete the dilution. The diluent washes the sample from the tubing so the system is ready for the next sample.
- Simple Dispensing — Solvent is drawn in from the reservoir and then dispensed out through the hand probe. Each fill is accompanied by a dispense. With the Basic controller it is not possible to perform multiple dispenses from a single fill of the syringe.
Many applications require more than the simple fill and dispense functionality provided by the Basic Controller. Unlock the full potential of the Microlab 600 by upgrading to the Advanced Controller. The Advanced Controller delivers these additional functions through intuitive Wizards or the Custom Method Editor. Wizard settings can be saved as Favorites or a Custom Methods on the 2GB Hardware Key.
For users with the Basic controller that would like to upgrade, it can be easily converted to the Advanced using a simple software kit (P/N 61500-02).