Incyte Arc cell density sensors come standard with Modbus RTU (RS485) digital protocol. The output signal can be used to communicate with ArcAir software on a PC or laptop or directly connected to process control systems that accept the digital signal. Like all other Arc intelligent sensors, Incyte Arc can also be connected through a Wi 2G Wireless Bluetooth adapter (Ref 243470). Examples of each wiring connection are provided below.
Incyte Arc Connection to PC
Incyte Arc Wiring to Process Control System (4-20mA and Modbus)
The addition of the Wi 2G Wireless Bluetooth Adapter adds wireless communication to the Incyte Arc sensor. In addition, the adapter enables two passive 4-20mA outputs. The Wi 2G Adapter also provides galvanic isolation for the analog outputs thus is a good choice for PLCs and other devices that lack their own isolation of the input circuit.
Incyte Arc Wiring to Process Control System (Modbus Only)
The simplest form of connecting Incyte Arc is using its native Modbus signal to communicate directly with the control system. In this case, the modbus registers must be setup in the control system to perform the proper read / write functions.
- Interested in Modbus programming for Incyte Arc? Refer to the Incyte Arc Manual for more details
- Curious about typical cell density measurement applications for Incyte Arc? Download our Applications eBook.
- Visit our Main Knowledge Base Article Page to see all related articles on cell density measurement.