Incyte Arc PCS Connections for Existing Cell Density Users
This article takes a look at different ways that the Incyte Arc cell density sensor can be connected to process control systems (PCS) and compares its methodology with the older, pre-amp based Incyte products.
4-20mA Analog Outputs
Incyte Arc requires the use of the Wi 2G Wireless Bluetooth Adapter (Ref 243470) to output 4-20mA signals. Two analog outputs can be transmitted from the sensor. These analog outputs can be individually configured for VCD (Viable Cell Density), Conductivity, or Temperature. Configuration is accomplished through ArcAir Software.
Pre-Amp versions of Incyte sensors require the ArcView Controller and the 4-20mA Analog Output Box (Ref 243850). The analog output box is a DIN rail mounted card that offers four outputs for cell density, and four outputs for conductivity. Setup and configuration is done through the Arc View Controller.
Modbus is the main output signal of the Incyte Arc sensor. It can be directly connected to the process control system or passed through the Wi 2G Bluetooth Adapter (Ref 243470).
Both Incyte Arc and the pre-amp version of Incyte offer Modbus digital protocol. The pre-amp version of Incyte always requires the Arc View Controller and Modbus is a standard output from the controller.
- Get in-depth on how Incyte Arc connects to your process with this article on wiring.
- Learn more about the measurement principles behind how cell density works.
- Visit our Main Knowledge Base Article Page to see all related articles on cell density measurement .