Range of the Stop Flow Rate is wrong
The Range of the Stop Flow Rate for several channels isn't correct. The wrong value we're appearing in the Liquid Class Editor: Liquid Details and Help. Unfortunately, a wrong value in VENUS one and two will be abort the method.
As a workaround, please use following values:
5 ml Channels: 10-2500 ul/s
384 CO-RE Head 50 ul: 1.5-240 ul/s